The SPiDeR Research programme
The research goal of SPiDeR is to further develop monolithic silicon active pixel sensors for future particle physics experiments and to demonstrate their viability for vertexing, tracking and calorimetry applications.
Monolithic silicon pixel sensors for particle physics - the scientific case
As particle physics colliders reach higher energies and luminosities, finely segmented pixel detectors will be needed for vertexing, tracking and even for electromagnetic calorimetry where high granularity has been shown to be important for jet resolution using particle flow algorithms (PFA). These algorithms require efficient separation of the jet into its constituent particles, with each particle being measured accurately in the detector with the best resolution. Specifically, charged particles are measured in a tracker, photons and electrons in an electromagnetic calorimeter and neutral hadrons in a hadron calorimeter. The critical issue for the PFA approach is then to correctly assign all the energy in the calorimeters to avoid double counting; this so-called "confusion" term can dominate the jet resolution obtained. The best approach to reduce such confusion is to design the detector specifically with PFA in mind. These requirements are nicely illustrated in Figure 1 which shows a simulated event. The challenge is to provide sufficient measurements along the path of a track so that it can be reconstructed and its momentum can be measured precisely. The high density of tracks close to the interaction region and again in the outer calorimeter region demand detectors with high granularity. The innermost tracking layers need few micron precision, for reconstruction of displaced vertices. In the intermediate tracking region the emphasis is on minimizing the total amount of material. This is quantified in the following paragraphs and summarised in Table 1.
The function of a vertex detector is to provide precision spatial measurements of tracks close to the primary interaction vertex without introducing too much material. The innermost layer is placed as close as possible to the interaction vertex (typically ~1 cm). To distinguish the tracks from displaced vertices the resolution needs to be a few microns, implying a pixel size of 20 μm or smaller and analogue readout. Typical occupancies through the bunch train are on the order of 100 hits per square mm or higher. Some form of time-stamping is required to get the occupancy per time-slice below ~ 1% as required for pattern recognition.
The radiation level expected at
Silicon technology is the only proven technique for realising the small pixel sizes required. In our proposal we investigate the use of a silicon sensor with in situ charge storage for time slicing, which has the potential to decrease the power and hence material budget for burst applications such as the cold superconducting linear collider. While not able to subdivide the much shorter bunch trains at CLIC , this is not needed due to the far lower luminosity per train. Readout spanning a number of CLIC bunch trains will deliver adequate background rejection.
The function of a tracking detector is to provide sufficient granularity and measurement precision to be able to find and reconstruct all charged particles, again without introducing too much material and to provide a precise measurement of the curvature. Achieving the required momentum resolution requires an rφ precision of ~15 μm. Pattern recognition for track reconstruction demands the detector occupancy to be less than 1%. This leads to two possible approaches - either long strips with time-stamping capability or higher granularity with no or limited time stamping. It is this second approach which we want to explore as it has the advantage that it can be used both in the barrel and endcap regions. The material per measurement layer is again a crucial parameter if multiple scattering and conversions are not to spoil the measurement which demands a minimal number of I/O lines and minimal power dissipation. The overall goal here is to minimise the material in front of the electromagnetic calorimeter. The radiation levels in the tracker are expected to be an order of magnitude lower than for the vertex region .
In this proposal we wish to investigate a novel solution which is an evolution of the ISIS and TPAC developments. Pixels will be realised using the 4T imaging technology which has recently become available to us and will provide better noise performance than previous 3T architectures like TPAC. It provides efficient charge collection and storage with low power. The readout will be realised as a conventional rolling shutter, with the columns subdivided to give the required time slicing. It is an important question for our R&D programme to determine whether this approach using a pinned photodiode or the ISIS approach using a photogate will provide full detection efficiency for minimum-ionising particles (MIPs) with compact clusters and adequate precision. This could be more challenging with the larger pixels (~25-50 μm) needed for tracking. The readout and memory buffers will be distributed over the pixel area with no dead space using the deep p-well INMAPS technology, which was pioneered for TPAC. The time-stamp resolution will be adjustable from the full 1 ms in the barrel down to 0.1 ms in the very forward region. Power dissipation even in the forward region is expected to be compatible with gas cooling. The radiation tolerance of the structure will be investigated to understand if the principle can be developed for other applications.
Digital electromagnetic calorimetry
Any foreseen future collider detector will need to reconstruct quark final states through hadronic jets. There has been an intense study of techniques to give good jet energy resolution over the last few years and the consensus is that Particle Flow Algorithms (PFA) are most likely to give the best performance. For electromagnetic calorimeters, silicon-tungsten (Si-W) sampling calorimeters are the favoured technology for a PFA-based calorimeter, as the tungsten absorber has a small radiation length (3.5mm) and Molière radius (9mm) while the silicon sensitive layers can be made very thin. Studies so far have used silicon detectors with diode pads with dimensions of several mm and have reconstructed the shower energy through an analogue measurement of the energy deposited in the silicon.
An alternative approach is instead to attempt to count the charged particles crossing the silicon detectors and use this as the shower energy estimator. Both the number of charged particles and the energy deposited in the silicon are on average proportional to the total energy of the electromagnetic shower. However, the energy deposited arises because of the passage of the charged particles through the silicon. The actual energy deposited is not a constant for each particle but depends on the path length and velocity of the particle, as well as having a fluctuation described by the Landau function. This means the energy deposited has extra effects which smear the value over and above the fundamental shower fluctuations which contribute to variations in the number of charged particles. This is illustrated in Figure 2, where an idealised simulation shows the fundamental shower resolution obtained when using the analogue energy deposited or the digital number of particles. Here, idealised means no instrumentation effects (electronics, etc.) is added to the analogue value and the number of particles is counted perfectly, so showing the fundamental differences of the two approaches. The basic ideal resolution of the analogue case is seen to be around 50% worse than for the digital case. Resolutions close to the ideal analogue value have been achieved with low noise systems. However it is not clear how close it will be possible to get to the ideal digital value in practise; to determine this is the major aim of this Workpackage.
Summary of the requirements for a pixellated silicon sensor at a linear collider
Table 1 Summary of the requirements for a pixellated CMOS sensor
Monolithic silicon pixel sensors - the technical case
The use of silicon sensors, including charge-coupled devices (CCDs) and hybrid silicon pixel detectors for vertexing, hybrid silicon strip detectors for tracking and silicon pad detectors for luminosity monitoring is by now well established in particle physics. Silicon is the technology of choice where high precision measurements are required and its use is likely to continue into the future, which is evident from the current proposals for sLHC upgrades and the recent linear collider detector concept studies. The urgent requirement is to develop novel solutions which have faster readout and/or lower power density so that the material per sense layer can be reduced by factors of 2-5. In addition the cost per unit area is a limiting factor.
In the early '90s CMOS Monolithic Active Pixel Sensors (MAPS) were proposed as imaging sensors. Charge-Coupled Devices (CCD) were then the dominant imaging technology but it was immediately recognised that an image sensor in CMOS has significant cost advantages as it allows the integration of the sensor with sophisticated readout electronics including analogue-to-digital conversion and signal processing. It also allows very small pixel sizes building on the reduction in feature size of CMOS. Since then the technology has continued to improve by the introduction of features such as low dark current diodes and charge transfer, which were previously found only in CCD technology. It was quickly shown that the devices can act as particle detectors and that they have the following attractive features:
Table 2 Architecture comparison
In order to investigate these issues we propose an R&D programme which addresses the fundamental issues discussed above by developing three sensors. Two of these devices are based on work initiated within the framework of the LCFI and CALICE-UK projects. The ISIS design will demonstrate low-noise charge storage in a pixel size which is compatible with the requirements of a vertex detector. The TPAC2 sensor will be a large area sensor which will be used to validate the principle of digital electromagnetic calorimetry. This should determine the limiting factors for operating a DECAL detector and so give an indication of the required future direction for development of such sensors.